Although nasal continuous positive airway pressure or non-invasive ventilation is used to manage some patients with acute lung injury due to COVID-19, such patients also demonstrate increased minute ventilation which makes it hard, if the device is used in line with the manufacturer’s instructions, to achieve adequate oxygen delivery. In addition, if a hospital contains many such patients, then it is possible that the oxygen requirements will exceed infrastructure capacity. Here we describe a simple modification of two exemplar ventilators normally used for domiciliary ventilation, which substantially increased the fraction of inspired oxygen (FiO2) delivered.
- non invasive ventilation
This article is made freely available for use in accordance with BMJ’s website terms and conditions for the duration of the covid-19 pandemic or until otherwise determined by BMJ. You may use, download and print the article for any lawful, non-commercial purpose (including text and data mining) provided that all copyright notices and trade marks are retained.https://bmj.com/coronavirus/usage
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.