Preproduction and current models of the miniature Wright peak flow meter have been compared with the standard Wright peak flow meter on normal and abnormal subjects. Early problems in production appear to have been overcome, and the current model agrees to within 3% with the standard peak flow meter, which is as close as the agreement between two standard instruments. The new mini-meter may be enclosed in a case, making direct comparisons with other instruments possible.
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.