Four cases are described in which totally cloth-covered Starr-Edwards valves (model 2300) had to be removed. All were causing significant haemolysis, two in the absence of a peripheral leak. The principal operative finding was destruction of the Dacron covering the struts. Following replacement of these prostheses there was complete resolution of signs and symptoms. The possible causes of haemolysis and consequences of cloth destruction are discussed.
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.